Are you a reseller or buying in bulk? Call us today.
Do you have questions about this product?
Get answers from real customers and in-house experts with AnswerBox.
Know the answer? Answer this question
Do you have a question about this product?
(based on 1 review)
Reviewed by 1 customer
Displaying review 1
Back to top
(2 of 2 customers found this review helpful)
Fluke going down the toilet?
from Aurora, IL
Comments about Fluke 971 Thermohygrometer:
I purchased one of these recently and was suprised to see it reading out of range as compared to other similar calibrated intruments in our lab. A replacement was ordered and it was within it's stated accuracy but the black scensor on the top of the unit was glued on at an angle, pretty sloppy. On another note, we send many of our Fluke instruments directly to Fluke for calibration and have had nothing but trouble. The list of issues is to extensive to list in this type of forum. I am currently looking for other brands and other calibration houses.
Bottom Line No, I would not recommend this to a friend
Was this review helpful? Yes / No
- You may also flag this review